Siemens products may only be used for the applications described in the catalog The SITRANS FUE energy calculator has been developed for Siemens. Fine Controls are the official Technology partner for the Siemens instrumentation product range type FUE, Inline Ultrasonic Energy Meters, In Stock. Dec 4, Siemens Industry Automation Division introduces the Sitrans FUE thermal energy calculator, designed for use in hot water, chilled water.
|Published (Last):||17 March 2018|
|PDF File Size:||1.64 Mb|
|ePub File Size:||2.64 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 50 Operation hardware 6.
Check cable plug connection of power supply module. Page 4 Note the following: Tue950 Procedures Waste industrial batteries are accepted back by Siemens or by the local Siemens representative. After installing main power cable, the inner plastic cover has to be installed with the two screws. Page 42 Tariff energy 2 1. Page 6 Power supply Communication is retained, e.
Operation hardware Sensors in forward cue950 return lines must fit right to bottom of sensor pockets. Carefully slide plug onto board. Technical Support Service and maintenance 7. Remove user sealing if present from integrator housing and open lid.
The display is also switched off, but can be switched on again by pressing any button. To print the manual feu950, please, download it.
Don’t show me this message again. Check both type plates for settings.
SITRANS FUE – Automation Technology US – Siemens
Short press Returns to menu 2. Slot 1 to the left is intended for the M-Bus or RS communication modules or the pulse input module. Siemens accepts no liability for amendment of legal metrological data if due950 seal has been broken. Please talk to your local Siemens contact or follow the return procedures of Siemens Flow Instruments described in chapter: Figure S – User sealing 3.
Thermal Energy Calculator
Ultrasonic flowmeters 7me portable flowmeter pages. Device installation, see Installation of device Page In case fke950 deviating environmental conditions, energy meter has to be remounted regularly for service. Sitrans Fue And Fus For connection of flowmeter, see flowmeter operating instructions.
Page 37 M-Bus or RS module is not used, pulse input module can also be installed in slot 1. In 1 Pulse 1 In 2 Pulse 2 counter counter 6. For connection of flowmeter, see flowmeter operating instructions.
Page 24 pipe by heating low temperature Calculator combined Forward “hot line” By heating: E – 9 Display Battery empty checked Monitors: Safety Notes Material compatibility Siemens Flow Instruments can provide assistance with the selection of wetted sensor parts.
RS output Type The optional plug-in RS communication module is a serial interface and permits data exchange with the calculator.
Short press Short press Short press fuf950 sec.
SITRANS FUE – Process Instrumentation – Siemens
Short press Returns to menu 3. Page 72 Further information www. Enter text from picture: